Emergency Stations - Pull String with auxiliary contact and LED options
E-114, E-114-1 and E-114-3 Emergency Stations feature satin finished stainless steel faceplates with silk-screened designations. They may be flush mounted on standard outlet boxes with single-gang rings, or surface mounted on Wiremold 5748 or equal with a depth requirement of 1 ¾“. Field wiring terminations are wire leads.
The E-114, E-114-1 and E-114-3 have DPDT contacts and pigtail wires. The E-114 has four; the E-114-1 has seven and the E-114-3 has eight pigtail wires.
Parts For Individual Replacement Units:
Pulling the nylon cord on CORNELL’s Emergency Station activates a call for help. The E-114-3 also has a LED light to confirm call placement. To cancel the call, slide the switch to the off position.
The satin finish stainless steel faceplate with silk-screened designations may be flush mounted on a standard outlet box with single-gang plaster ring, or surface mounted on a Wiremold 5748 or equivalent.
The contractor shall furnish and install these CORNELL Emergency Stations as indicated on the plans. Faceplates shall be satin-finish stainless steel with silk-screened designations. Calls in progress can be cancelled by moving the switch to off position. (A red LED on the E-114-3 indicates calls in progress). Stations shall be mounted on standard outlet boxes with single-gang rings or Wiremold 5748 or equivalent.
- Power Requirements: 24VDC
- Operating Environment: 50-120°F Indoor Non-condensing
- Physical Faceplate Dimensions: 4 ½ ”H x 2 ¾ ”W (Standard 1 Gang)
- Mounting: Single-Gang back box with 1¾” minimum depth
- Wiring: Field wiring terminations are wire leads.
Get exactly the system you want.
Audio and visual nurse call systems for assisted living facilities, nursing homes, hospitals and outpatient medical clinics.
Traditional lights, buttons and pull cords or smartphone/WiFi integrated. Let us know what you’re trying to do and we’ll take care of you.