Get the system you need.
Audio and visual nurse call systems for assisted living facilities, nursing homes, hospitals and outpatient medical clinics.
Traditional lights, buttons and pull cords are smartphone/WiFi integrated. Let us know what you’re trying to do and we’ll take care of you.
Order by Noon CST for possible same day shipping*
*See Terms & Conditions
Emergency Stations with Push Button or Drop Flag Actuation
Order Replacement Parts
|Part E-101||Part E-113|
E-103/E-113 stations feature satin finish stainless steel faceplates with silk-screened designations. They may be flush mounted on Wiremold 5748 or equal, with a depth requirement of 1 ¾”.
- E-103: May be used with dual status systems. Field wiring terminations are with screw terminals for maximum 16-guage wire.
- E-113: For single status only. Wire lead terminations.
E-101: Push Button Emergency Station features a satin finish stainless steel faceplate with silk-screened designations. Stations can be used on single or dual status systems. They may be flush mounted on a standard outlet box with single gang rings, or surface mounted on Wiremold 5748 or equal with a depth requirement of 1 ¾". Incandescent lamps not acceptable. Field wiring terminations are with screw terminals for maximum 16-gauge wire.
E-103/E-113: A touch of the DPDT switch button on the UL listed E-103/E-113 Emergency Station initiates an emergency call, verified by mechanical flag drop. Calls may by cancelled by pushing the button again.
E-101: A touch of the push button type switch on the E-101 Emergency station activated a call for help, verified by a red LED indicator. Calls can be cancelled by pushing the nurse call button again.
The satin finish stainless steel faceplate with silk-screened designations may be flush mounted on a standard outlet box with single-gang plaster ring, or surface mounted on a Wiremold 5748 or equivalent.
The contractor shall furnish and install CORNELL Push Button Emergency Stations as indicated on the plans. Faceplates shall be satin finish stainless steel with silk-screened designations. Calls in progress (E-101) are indicated by a red LED. Calls in progress (E-103/E-113) are indicated by mechanical flag drop. Calls may be cancelled by pushing the button again. Stations shall be mounted on standard outlet boxes with single-gang rings or Wiremold 5748 or equal.
- Power Requirements: 24VDC
- Operating Environment: 50-120°F Indoor Non-condensing
- Physical Faceplate Dimensions: 4 ½ ”H x 2 ¾ ”W (Standard 1 Gang)
- Mounting: Single-Gang back box with 1¾” minimum depth
- Wiring: #22 AWG Minimum, #16 AWG Maximum