Get the system you need.
Audio and visual nurse call systems for assisted living facilities, nursing homes, hospitals and outpatient medical clinics.
Traditional lights, buttons and pull cords are smartphone/WiFi integrated. Let us know what you’re trying to do and we’ll take care of you.
Order by Noon CST for possible same day shipping*
*See Terms & Conditions
Need Some Assistance?System Quote System Design/Installation 1-800-558-8957
|Part D-111||Part D-112|
(D-111) The Cornell D-111 Duty Station includes an on/off tone switch that can be passed, green LED to indicate normal call, and red LED to indicate emergency call.
(D-112) The Cornell D-112 Duty Station includes a green LED to indicate normal call, and red LED to indicate emergency call. It is also furnished with an adjustable volume pot that is preset upon installation.
(D-111) The D-111 Duty Station has a tone on-off switch that is provided to silence the audible alert by staff. Normal calls are indicated with a steady green LED and slow intermittent tone. Emergency calls are indicated with flashing red LED and rapid tone.Operation:
(D-112) The D-112 Duty Station indicates normal calls with steady green LED and slow intermittent tone. Emergency calls are indicated with flashing red LED and rapid tone.
The satin finish stainless steel faceplate with silk-screened designations may be flush mounted on a standard outlet box with single-gang plaster ring, or surface mounted on a Wiremold 5748 or equivalent. Field wiring terminations are with screw clamps, maximum 16 gauge.
The contractor shall furnish and install CORNELL Duty Stations (D-111) or (D-112) as indicated on the plans. Flashing red LED and rapid intermittent tone shall indicate an emergency call; steady green LED and slow tone shall indicate a normal call. Incandescent lamps are not acceptable. The station shall fit on a standard single-gang plaster ring or Wiremold 5748 surface box or equivalent, with a minimum depth of 1 ¾” required. Terminations for field wiring shall be with screw clamps.
- Power Requirements: 24VDC
- Operating Environment: 50-120°F Indoor Non-condensing
- Physical Faceplate Dimensions: 4 ½ ”H x 2 ¾ ”W (Standard 1 Gang)
- Mounting: Single-Gang back box with 1¾” minimum depth
- Wiring: #22 AWG Minimum, #16 AWG Maximum
Download Brochure & Wiring Diagram