Get the system you need.
Audio and visual nurse call systems for assisted living facilities, nursing homes, hospitals and outpatient medical clinics.
Traditional lights, buttons and pull cords are smartphone/WiFi integrated. Let us know what you’re trying to do and we’ll take care of you.
Order by Noon CST for possible same day shipping*
*See Terms & Conditions
Corridor Light with Pulsing Tone
The LS-101AP provides visual and audio notification when activated. Each device includes an incandescent bulb and piezoelectric type sounder, mounted on either a single or double gang stainless steel plate, with a dome cover. The device may be mounted in any position. The dome cover is illuminated by the incandescent bulb.
Choose from a variety of bright, long-life LED modules in various colors, sold separately in White (8020-001W), Red (8020-001R), Blue (8020-001B), Green (8020-001G), or Amber (8020-001A). Domes can also be silkscreened with custom designations. Please call Cornell for pricing and availability.
When an associated call station is activated, the bulb illuminates the dome cover and will sound the tone device. The light and tone generated will be active until the station is manually reset.
Dome covers are easily removed from the front for simplified bulb replacement. Only use Cornell approved replacement parts.
All models mount on standard, single or double gang boxes or plaster rings, with a required depth of 1 inch.
The contractor shall furnish and install the CORNELL LS-101A or an LS-201A series call indicator corridor lights as indicated. Faceplates shall be stainless steel with mounting screws. Calls shall be indicated by an incandescent bulb covered by a translucent plastic cover and a piezoelectric tone device mounted within the cover only on a single gang. The light socket and sounder each have 2 leads, a positive and negative. The sound output shall be 90 dB at 24 VDC.
• Power Requirements: 24VDC
• Operating Environment: 50-120°F Indoor Non-condensing
• Physical Faceplate Dimensions: 4.5” H x 2.75” W (1 Gang)
• Mounting: Single or double gang back box with 1” minimum depth
• Wiring: #22 AWG Minimum