Get the system you need.
Audio and visual nurse call systems for assisted living facilities, nursing homes, hospitals and outpatient medical clinics.
Traditional lights, buttons and pull cords are smartphone/WiFi integrated. Let us know what you’re trying to do and we’ll take care of you.
Order by Noon CST for possible same day shipping*
*See Terms & Conditions
Need Some Assistance?System Quote System Design/Installation 1-800-558-8957
Corridor Light, Two Lamp on Single Gang Plate
Light Notification System for Nurses
The L-102 corridor light notification system is an individual lamp mounted on a single-gang plate.
This corridor light notification system is ideal for nursing homes, senior care facilities, schools and more. Residents activate the device or bedside station to illuminate the system to notify staff of an emergency.
Available with one, two, three, four or six lamps and full range of optional silk-screened designations. Cornell Corridor Lights feature translucent wedge-shaped covers in red or white.
The light will illuminate when associated station is activated.
Dome covers are easily removed from the front, for simplified lamp replacement.
Corridor lights can be made in 1-6 light configurations with white or red domes. Lights can be made red, blue, amber, yellow, or green with additional filters. Domes can also be silk-screened.
Mount on standard, single-gang boxes or plaster rings, with a required depth of 1 inch.
The contractor shall furnish and install the Cornell Series 4000 Corridor Lights with two lamps. The domes shall be (red) or (white), with optional designations silk-screened on the surface. The stainless steel faceplate shall mount on a standard, sing-gang electrical wall box or single-gang plaster ring, with external connection achieved via pigtails. The dome covers shall be removable from the front for easy replacement of the lamps.
- Power requirements: 24VDC
- Operating Environment: 50-120F indoor non-condensing
- Physical Faceplate Dimensions: 4 1/2"H x 2 3/4"W (1 gang)
- Mounting: Single-gang back box with 1" minimum depth
- Wiring: #22 AWG Minimum